PLUS LINE: The standing aids from the EASY UP Series help seek the upright position actively, as they have a support structure that accompanies the user throughout the standing sequence. Standing up becomes a lot easier for users with poor motor control.
the tray, trunk support and side handles help the user throughout the standing sequence. The EASY UP support elements move parallel to the plane while the person comes to standing.
ADJUSTING THE STRUCTURE IN HEIGHT
the support structure adapts to the user’s height easily and with no effort as it is operated electrically or by gas strut. This adjustment can also be made by the user in a quick and easy way and with no need for external help even when they are already standing. The knee, lateral and trunk supports are adjusted accordingly, for greater safety and comfort of the user.
Access to the aid is made easier and safer by the alignment of the seat with the original seating position. When transferring from the wheelchair to the standing aid, the seat, unlike the old model, can always remain horizontal and align to the user’s seating level by adjusting the aid height.
The sliding footplates, which are independent and adjustable in depth, reduce the distance between the person in the wheelchair — or sitting — and the base of the structure. This helps transfer the user’s feet onto the standing aid. The footplates can be adjusted even when the user is on them.
1 - ADJUSTING THE STRUCTURE IN HEIGHT:
the support structure adapts to the user’s height easily and with no effort as it is operated electrically or by gas strut. This adjustment can also be made by the user in a quick and easy way, with no need for external help even when they are already standing. The knee, lateral and trunk supports are adjusted accordingly, for greater safety and comfort of the user.
Height adjustment: 200 mm
Seat minimum height: + 380 mm
Seat maximum height: + 580 mm
1 - Adjusting the support structure in height also enables a similar adjustment of the shin rests, side handles and tray at the same time.
2 - The aid is suitable for users who are between 140 and 200 cm tall and weighing up to 140 kg.
2 - ADJUSTING THE FOOTPLATES
All upright standers feature two footplates at the base of the frame. These slide independently from each other
so they can be adjusted to meet the wheelchair users’ accessibility requirements and improve their posture.
Depth adjustment: 100 mm
3 - ADJUSTING THE HEEL RESTS
Independent depth adjustment for each heel rest. A special accessory strap (ref. code AC0047) can be added to
safely position the feet in the footplate.
Width: 90 mm x 2
Center distance: minimum 206 mm; maximum 386 mm
4 - ADJUSTING THE SHIN RESTS
The soft and wrap-around shin rests allow the knees to move freely while the user comes to standing.
The possible adjustments are as follows:
Width: 170 mm; Center distance: min. 210 mm; max 380 mm (independent)
Depth: 100 mm (independent)
Height adjustment: 40 mm manually, further 200 mm by servo-adjustment
Rotation: +/- 20° (independent)
5 - ADJUSTING THE SEAT
Depth is adjustable in standard seat models. A visual indicator signals the maximum safe adjustment allowed.
It is possible to slide out the seat completely if needed.
Depth: 120 mm
6 - ADJUSTING THE SIDE HANDLES
The side handles are adjustable in width and can be reversed to better restrain the trunk if need be.
Width 110 mm x 2
Center distance: minimum 360 mm; maximum 580 mm
Handles reversal: allowed
7 - ADJUSTING THE TRAY
The tray tilt can be adjusted directly, whereas the table height adjustment is achieved by adjusting the support structure. Finally, the thoracic support built in this table is adjustable in depth.
Tilt adjustment: 5° x 4 = 20°
Depth adjustment: 80 mm